Disconnect the battery cable from the main board. Follow the Touch Panel Calibration screen instructions using a stylus. The screw is beneath the manifold. Apply detergent to the product using a soft cloth. Start display at page:.
|Date Added:||24 June 2007|
|File Size:||11.22 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Refer to the cleaning instructions from the sensor manufacturer.
Cut and discard the tie wrap holding the valve. To make this website work, we log user data and share it with processors.
The Fairbanks scale is self-zeroing.
Midmark 9A629001 Iqvitals Table Mount With Usb And Serial Cables
Pull firmly on the pump to remove it from the adhesive stabilizer pad and main board. Device is not powered on.
Readings seem low or high. It may also be used to alert against unsafe practices Equipment Alert Indicates a potentially hazardous situation which could result in equipment damage if not avoided. Trademark Information is a registered trademark of More information.
Be sure that the wire coils back next to the pump and is also secured by the tie wraps. Class B devices have a Federal Communications More information.
Battery is fully discharged. Remove the detergent from the product using a cloth dampened in distilled water.
IQvitals and IQvitals PC
Connect the battery bua More information. Attempting to do so may permanently damage the equipment. Speaker Cable red and black Temp Cable black and white Install: This document applies to Part Numbers: If pressure is not usg before mmhg, check the accuracy of the test fixture.
Copy to Hard Drive B Do not put the lithium ion battery in a refuse container. The iqvittals sticker will need to be replaced if extra port is exposed see photo below. It is important More information.
Pull firmly on the valve.
IQvitals USB (Driver Removal) – Download
The blood pressure cuff should be at the same level as the patient s heart. The battery should last for years under normal use and can be replaced via a dedicated access door on iqvitxls back of the device. Caution In case of malfunction, call the Midmark Support Services department atoption 2, and be prepared to describe the problem.
Instruct patient to remain still. Nicholas Carpenter 1 years ago Views: Server Safety Information To reduce the risk of bodily injury, electrical shock, fire, and equipment damage, read this document and observe all warnings and precautions in this guide before installing More information. Trademark Information is a registered trademark of. Unplug the iqvital and temperature cables from the current I-O board. Install all screws in new rear cover. Remove the input panel and install it on the new main board by unscrewing the two screws, removing the input panel, and screwing the input panel onto the new main board.
Do not over tighten screws, but make iqvitlas that the brackets do not move after installation. C and C Series.